Powerpoint template for scientific posters (Swarthmore College)
Tracheostomy Tube Security Device
Rebecca Clayman, Paul Fossum, Katie Pollock, Kim Safarik
Department of Biomedical Engineering
Advisor: Paul Thompson Client: Dr. Timothy McCulloch, Department of Otolaryngology
Final Design and Testing
In patients who have had a tracheotomy performed, a major post surgery
problem is discomfort from the tracheotomy collar. The collar must be secured
tightly to keep the tracheotomy tube in place so the patient can successfully
breathe. However, if the collar is kept at too high of a tension, it can result in
ulcers on the patients skin. Our goal is to design a tracheotomy strap that is
comfortable, easy to clean and equipped with monitoring devices to ensure
that proper pressure is maintained on the neck.
A red LED light is attached to a
battery in series. Connections to
the other side of the battery and
light rest on a thin piece of plastic.
As the tracheotomy strap becomes
looser or tighter, it moves which
causes the leads to make contact
with the foil attached to the strap
and complete the circuit, turning
on the light. This will alert the
nurse on duty of a potential
problem when she checks on the
Easy to clean/maintain
Monitoring system to ensure proper
Material to adequately
expand/Contract with neck
5-7 day use
Ulcers and Banding
Caused by excess pressure
Loose strap = loose tube
Readily absorbs moisture
Jersey mesh material
Belt holes and pin
Fits neck size approximately
11 to 20 inches in circumference
Equal area sections of strap were
allowed to soak in 10mL of water
for 1 min. The straps were
removed and the volume of
remaining water was measured.
The volume difference was taken
as the volume absorbed.
Foam (Blue strap): 0.3014mL/cm2
Mesh (Our strap): 0.0274mL/cm2
We polled an unbiased, random
sample of people on their opinion
of comfort, safety and function
while wearing three straps: the
cotton strap currently used by our
client(1), the blue Velcro strap
used by many of his
colleagues(2), and the strap that
Our strap was ranked best in
overall comfort, tied for least
abrasive, ranked highest in safety,
and highest in overall function
according to our subjects.
Final Design Strap Features
Breathable/Non-absorbent mesh material
Clinical feedback and testing
Different size necks; child/infant
and bariatric straps
Professor Paul Thompson
Timothy McCulloch, M.D.
Chairmen of Otolaryngology, University of Wisconsin School
of Medicine of Public Health
Anna Krupp, Clinical Nurse Specialist
Department of Biomedical Engineering
"The Essence of Love" Two Things: Love is a Concept Abstract Types of Definitions: Real Nominal Operational Error/Bias Composite Measure Index Scale Index Sum or Average of Scores Political Involvement Political Involvement Scale Unidimensionality Same Score Different Responses Assign Scores...
Pohyb částeček kapalin objevil anglický botanik Robert Brown ( 1773 -1858) při pozorování vzorku zředěného mléka s pylovými zrny. Zrna se nárazy částic kapalin pohybovala neuspořádaným pohybem. Brownův pohyb Vlastnosti kapalných látek Vlastnosti plynných látek Plynné látky nemají pevný tvar...
Retrieved on November 17, 2010, from InfoTrac Student Edition. The history of the question of authorship is outlined. This article aims to authenticate Shakespeare's authorship and claims any evidence to the contrary is a delusion. Kastan, D. S. (2008). "To...
Ethics is the study of moral obligation, or separating right from wrong. Ethics converts values into action. Unethical acts can be legal or illegal. Moral intensity is the magnitude of an unethical act, such as using company jet for a...
Terms were 3/15, n/45, FOB shipping point. 20. Paid freight charges, $300. Jul. 4. Returned $900 of inventory to Surry Diamonds. 14. Paid Surry Diamonds, less return. 16. Purchased inventory of $3,800 on account from Swift Diamonds, a jewelry importer....
More Tips Body Language Posture Be confident and open Movement Don't be a rock don't be a rabbit Hands/arms Hold a marker Use gestures Eye contact Interaction with audience Do what feels natural Voice R.S.V.P. Rhythm Speed Volume Pitch Make...
A comparison is made with the current Av formulae of R55, a proposal from WAP and the New Zeeland standard NZS 5446:2007. Those results are compared to some measured results from VBG. Further it is considered that there have been...
Ready to download the document? Go ahead and hit continue!